Writer： admin Time：2023-02-07 15:37:06 Browse：117℃
Store the cap ：
Regular electrode maintenance requires cap to be stored ina protective cover with a damp sponge and checked and watered regularly, so as to keep the fluorescent film in a moist state for a long time.If the sensor fluorescent caphead is dry for a long time, the measurement results will drift, and it needs to be soaked in water for 48 hours before continuing to work.
Yosemitech Technologies Co., Ltd
Add: Bldg,25,CECEP Industrial Park, No. 18 Dongchang Rd. Suzhou Industrial Park, Jiangsu Province,China 215126, China